*Note: This page contains materials in the Portable Document Format (PDF). The free Adobe Reader may be required to view these files.
Improperly stored vaccines result in wasted vaccine, and in some cases, suspension of providers from the VFC Program. When VFC Program providers do not maintain the recommended refrigerator/freezer temperatures, vaccines may lose their potency and result in children being inadequately immunized and required to get additional vaccinations.
Vaccine Storage Equipment
Providers must have appropriate vaccine storage equipment that can maintain proper temperature conditions:
- Alarm-equipped, stand alone refrigerator and freezer.
- Refrigerator unit must maintain temperatures between 35° to 46°F (2° and 8°C) year-round.
- Separate freezer compartment with a separate exterior door.
- Freezer compartment should maintain an average temperature of 5°F (-15°C) or colder year-round.
- Combination refrigerator/freezer units sold for home use are acceptable for vaccine storage.
- Refrigerators and freezers must be large enough to hold the year's largest inventory.
- Refrigerators and freezers must have a working thermometer certified in accordance with the National Institute of Standards and Technology (NIST) or the American Society for Testing and Materials (ASTM) standards placed in a central area inside each storage compartment.
- Refrigerators and freezers must be dedicated to the storage of vaccines. Food and beverages must not be stored in a vaccine refrigerator or freezer.
Temperature monitoring should be the responsibility of the primary and back-up vaccine coordinators. If other staff must monitor temperatures, train them on how to respond to and document actions taken when temperatures are outside the appropriate range.
- Post a Fahrenheit or Celsius Temperature Log for Vaccines on the storage unit door or in a readily accessible place.
- Record refrigerator and freezer temperatures in the Temperature Log for Vaccines twice a day (beginning and end)
- Ensure that refrigerator temperatures are between 35° to 46°F (2° and 8°C).
- Keep the freezer temperature at 5°F or lower (-15°C or colder).
- Monitoring and recording temperature twice a day is required, even if a continuous graphing/recording thermometer or a digital data logger is used.
- Correct improper vaccine storage conditions immediately, including inappropriate exposure to light and storage temperatures outside the recommended ranges.
- Maintain and store the completed Temperature Log for Vaccines for 3 years. The Immunization Section field staff will review this form when they conduct on-site visits.
- Monitor vaccine storage temperatures by using a minimum/maximum thermometer or continuous recording thermometer in the refrigerator and freezer. Follow the manufacturer's recommended schedule for recalibration of the certified thermometers.
Back to Top
Because of the critical importance of administering safe and effective vaccines, we cannot guarantee continued participation in the Florida VFC Program for providers who do not adhere to these requirements. If it is determined the vaccine is stored improperly and is not usable during an onsite review, providers will be instructed to return it to McKesson Specialty, the centralized national vaccine distributor, immediately. Vaccine shipments will be suspended until the provider meets the proper requirements. Be sure to:
- Familiarize the appropriate staff with vaccine storage and handling requirements.
- Post the temperature log on the storage unit door or a readily accessible place.
- Record refrigerator and freezer temperatures twice daily, as indicated on the log.
- Adjust the thermostat if temperature is recorded outside the recommended range and contact the vaccine manufacturer for further instructions.